- 1 Terms in Engineering Measurements
- 2 Calibration:
- 3 Sensitivity:
- 4 Readability:
- 5 True size:
- 6 Actual size:
- 7 Hysteresis:
- 8 Range:
- 9 Span:
- 10 Resolution:
- 11 Dead Zone:
- 12 Threshold:
- 13 Backlash:
- 14 Response Time:
- 15 Repeatability:
- 16 Bias:
- 17 Magnification:
- 18 Drift:
- 19 Reproducibility:
- 20 Uncertainty:
- 21 Traceability:
- 22 Parallax:
Terms in Engineering Measurements
The following terms are used in engineering measurements and have their own definitions.
If a known input is given to the measurement system the output deviates from the given input, the corrections are made in the instrument and then the output is measured. This process is called “Calibration”.
Sensitivity is the ratio of change in the output signal to the change in the input signal.
Refers to the ease with which the readings of a measuring instrument can be read.
The theoretical size of a dimension that is free from errors.
Size is obtained through measurement with permissible error.
All the energy put into the stressed component when loaded is not recovered upon unloading. so the output of measurement partially depends on an input called Hysteresis.
The physical variables are measured between two values. One is the higher calibration value Hc and the other is the Lower value Lc.
The algebraic difference between higher calibration values to lower calibration values.
The minimum value of the input signal is required to cause an appreciable change in the output known as resolution.
It is the largest change in the physical variable to which the measuring instrument does not respond.
The minimum value of the input signal that is required to make a change or start from zero.
The maximum distance through which one part of the instrument is moved without disturbing the other part.
The time at which the instrument begins its response to a change in the measured quantity.
The ability of the measuring instrument to repeat the same results during the act measurements of the same quantity is known as repeatability.
It is a characteristic of a measure or measuring instrument to give indications of the value of a measured quantity for which the average value differs from the true value.
It means the magnitude of the output signal of the measuring instrument many times increases to make it more readable.
If an instrument does not reproduce the same reading at different times of measurement for the same input signal, it is said to be measurement drift.
It is the consistency of the pattern of variation in the measurement. When individual measurements are carried out the closeness of the agreement between the results of measurements of the same quantity.
It is nothing to establish a calibration by step-by-step comparison with better standards.
There is an apparent change in the position of the index relative is to the scale marks.
- Automobile Air Suspension System, 3 Basic Components, and Working
- Hydraulic Brake System Of An Automobile | Components, Construction, And Working Of Hydraulic Braking System
- Drum Brakes a Type of Mechanical Brake | Construction and Working of a Drum Brake System
- Lighting System of a Car | Car Lightings
- Pneumatic Braking System In Automobile | Air Braking System | Construction And Working Of Pneumatic Braking System
- Torsion Bar Suspension System In Automobile | Construction And Working Of Torsion Bar Suspension